Definition of device
Devices are improved in a way of making mild and cold sterilization by using disposable refills that contain 0 ethylene oxide gas.
For one sterilization, one reliff that is 100 gm and in the model of ETOSTD 135. Sterilization process ocur in a temperature of 37°C, 45°C, 55°C or 63°C. This choice is made by the user through the touch buttons on the control panel. In addition to this, except for the above-mentioned standart values, sterilization temperature and time have an option of ‘user-defined program’ that can be scheduled by the user.
Thanks to this option, there is a possibility of making sterilization between any temperature of 33°C ile 65°C.
Sterilization cabin is at prismatic structure and its volume is 135 liter for EOSTD 135. Its cabin dimension is 45 cm X 45 cm X 67 cm ( height X widthX depth) for ETOSTD 135.
In the ETO_KRT serial devices, sterilization cabin can also be used as air-conditioning cabin.
Devices carry out ethylene oxide sterilization at 4-phase. Respectively these phases are: CONDITIONING PHASE, STERILIZATİON PHASE( expousure), GAS DISCHARGE PHASE (exhaust) and AİR-CONDITIONING PHASE.
In user defined programs, temperature can be adjusted any of the intended rating between 33°C and 65°C.
Time of the air-conditioning phase can be chosen by the user between the hours of 0-99. Air-conditioning process is realized washing by mild air, for the purpose of diffusing a low grade of gas completely and in a short time that may be in the packaging of the materials sterilized.
The pressure, moisture, tempereture conditions and functional process required for sterilization is realized completely and otomatically by an electronic data processing unit and kept under control during the sterilization. All functional processes of device and error massages are written down with the help of a recorder.
Devices that having the same charecteristic features and different dimensions can be produced on order. Please contact us for more information.