SonoFly-3000 unit for ultrasonic diagnostics is a fixed ultrasonic system of visualization, processing, archiving and documenting of information. Ultrasonic information can be presented in the real-time mode, freeze mode and cineloop mode. The possible modes of information reproduction are as follows: B, B+B, B+M, M. Wide choice of cheap electronic probes contributes to the universality of systems application (abdominal investigations, obstetrics, gynaecology, pedonosology, cardiology etc.).
The software installed in the system allows the user easy control of the system operating modes, measurement of the object sizes, time characteristics, realization of the necessary calculations, completing of the examination reports, databases, archives and computer atlases. The images can be sent from the system both by local networks of the medical establishment and Internet.
In Scan Mode:
Imaging modes B, B+B, B+M, M.
Amount of simultaneously connected probes Two. Program switching.
Max Scan Depth 90-220mm. Depends on probe type.
Imaging Shift from Probe Surface Up to 40mm. Depends on probe type and scan mode.
Frame Average 0% - 90%.
Zoom in Image Area Fragment 125x125pix. 4 times enlargement. Colour contrast of the enlarged area. Possibility of measurements in the enlarged area.
Image Brightness & Contrast
* Time Gain Control in 5 depth zones.
* Changeable .
* Continuously adjustable dynamic range.
* Adjustable radiation power.
Focusing One/two focal zones with continuously changeable focus zones.
Scan Density 128 or 189 beams per frame. Program switching.
Image Orientation. Program adjustment LEFT/RIGHT, UP/DOWN.
M-Mode Specifications Adjustable rate of M-scanning in M and B+M modes (200, 100, 50 lines/sec).
M-line location choice.
High-Pass Filter Personal preset for each mode.
Cineloop Last 128 cadres record. Ability of reproduction, frame-accurate drop, record to hard disc as a reel.
In Freeze Mode:
Measurements B, B+B Modes:
* linear dimensions;
* Sectional area by ellipsis method;
* Elliptic fibration perimeter;
* Irregular-shaped sectional area (inking);
M, B+M Modes:
* rate of movement;
* axical displacement (on the axle perpendicular to the probe surface);
Images Record. Up to ten each scan session.
Measurement Results Report With possibility of naming (, etc.) and subsequent automatic inclusion into report text.
Automatic Calculations Changeable obstetrics standard, calculation of major indexes, comparison to standards, inclusion of results into report text.
Additional Colour contrasting indicators of the details of the image, body markers with the probe distribution in the investigated part of the body.