The Electrically Operated Compression Testing Machine is of rated capacity is fixed to the base. The upper platen has got a self-aligning action and is attached to a screw, which passed through the crosshead plate, and can be raised or lowered for initial clearance adjustment. The lower platen rests on the jack ram and is positioned with the help of a centering pin. Loading is accomplished by upward movement of lower platen. A dust cover is provided on the jack to prevent any dust from going into the cylinder. A spacer with a centering locating pin is provided to test small specimens. The lower and upper platens of the machine are hardened ground and polished. Guidelines are marked on the lower platen to keep the specimen coaxial with the hydraulic jack.
- Spare Oil Seal
- Brick Testing Attachment
- This test measures the compressive strength of bricks to be used in construction for load bearing walls. Specimens are first sampled from batches in a standard manner. The bearing faces are then either made perfectly flat by capping with sulphur or soft wooden spacers are used to take up surface defects, this will avoid any point loading. If frogs are present (indentations in the face) then these are filled with standard mortar. The maximum load sustained by the bricks before failure is then measured and the compressive strength calculated.
- Flexural Test Attachment
- The test method essentially involves applying a load at the centre of a beam of concrete supported at its ends. The load required to break the specimen is then recorded.
- Cube moulds of various sizes
- Accelerated curing tank
- Hydraulic oil (TIIN of 5 liters)